|Storage:||At -20±5℃, Protected From Light||Applicable:||Micgene 242 / 244 、ABI QuantStudio5|
|Mini Limit:||10 Copies/μL||CV:||≤5%|
Taq Pcr Kit Edwardsiella Piscicida Septicaemia Virus,
Edwardsiella Piscicida Septicaemia Virus DNA Test Kit,
EDW.P Taq Pcr Kit
Edwardsiella piscicida septicaemia virus (EDW.P) Nucleic Acid DNA Taq polymerase PCR kit
This kit is suitable for the specific detection of pathogenic Edwardsiella piscicida septicaemia virus (EDW.P) in fish viscera tissue samples.
This kit selection Edwardsiella piscicida septicaemia virus (Edw) P) specific fragment to design primers probe probe, the probe can with primer amplification area in the middle of a specific binding DNA template, in the process of PCR extension, the circumscribed Taq polymerase enzymes from the 5 'end of fluorescent groups on the probe will be cut down, make it free in the reaction system, thereby quenching groups out of the 3' end, Fluorescence was emitted and detected by fluorescence PCR instrument to achieve the detection of Edwardsiella piscicida septicaemia virus (EDW.P) in the completely closed reaction system.
|PCR reaction liquid||24 tubes|
|Positive control||1 tube|
|Negative control||1 tube|
1. Sample type
Fish liver, intestine, stomach and other pathological tissues.
2. Sample collection and transportation
Samples shall be collected and transported in accordance with SC/T 7103. Samples shall be transported to the laboratory as soon as possible under refrigerated conditions to avoid repeated freeze-thaw.
3. Sample preservation
It can be stored for 24 hours at 2℃~8℃, and can be stored for a long time at -20℃.
PCR amplification and fluorescence detection (PCR detection area):
Put each reaction tube into the PCR instrument and perform PCR amplification according to the following procedures:
|60℃||30s collect fluorescent|
FAM is selected for detection channels.
The results of interpretation:
|Negative||No Ct value or Ct value≥45|
Note: The result is judged to be grey area, which needs to be rechecked. If the recheck result is Ct value < 45 is considered as positive, and no Ct value or Ct value ≥45 is considered as negative.
Product performance index:
The positive reference materials of testing enterprises were all positive.
2. Minimum detection limit: the detection limit of this kit is 10copies/μL.
3. Precision: Coefficient of variation (CV,%) of intra-batch precision Ct value is ≤5%.
4. Linear, linear correlation coefficient | r | >0.980.
Operation details please check from IFU!
Contact Person: Ms. Lisa