|LOD:||2 PPB (µg/kg)||Warranty:||12 Months|
|Packing:||10T / Box||Specimen:||Pork, Chicken, Duck, Fish, And Shrimp|
120uL Food Safety Rapid Test Kit OFL,
Food Safety Rapid Test Kit Ofloxacin,
Ofloxacin Diagnostic Kit Colloidal Gold
Ofloxacin lateral flow assay diagnostics rapid test card tissue, Aquatic products
Ofloxacin (OFL) rapid detection card is convenient, fast and sensitive, suitable for on-site mass sample detection. The test card contains an antigen in the test area (T) and an antibody ⅱ in the control area (C) which are fixed in advance in the nitrocellulose membrane. After the sample was dissolved, the gold-labeled antibody in the micropores was incubated for 3 min at room temperature, and the gold-labeled microporous solution was added to the detection card and sample Wells for 5-8 min reaction. According to T area, C area color contrast to determine the result. Whether ofloxacin was present or not, a purplish red band appeared in region C.
Pork, chicken, duck, fish, and shrimp: 2 PPB (µg/kg)
|1||Ofloxacin rapid test card|
|6||Sample Reconstitution fluid|
Preservation and stability:
Store in dry place at 20±5℃, do not freeze, avoid direct sunlight.
This test card is valid for 12 months. The production date is shown in the package
Materials required but not provided:
|4||Nitrogen (air) blower|
1. Restore the test card and sample solution to room temperature before use;
2. Open the aluminum foil bag, take the required amount of gold label microhole box detection card, mark and put it on the table (note: after removing the microhole, seal the aluminum foil bag immediately to prevent moisture);
3. Absorb 120 μL of the sample to be tested into the gold label microhole, blow carefully until the purplish red particles at the bottom of the hole are completely dissolved (note: blow gently to avoid bubbles), and incubate at room temperature for 3 min.
4. Add all the solution in the gold label microhole to the detection card and sample well for reaction of 5-8 min and then interpret the result.
Negative: C line color, T line darker than C line color or consistent,The test results were negative.
Positive: C line is colored, T line is lighter than C line or T line is light green, and the test result is positive.
Invalid: Line C does not display color, regardless of whether line T displays color, the detection card is judged to be invalid.
Contact Person: Ms. Lisa