|Package:||48 Tests/kit||Expiration Date:||12 Months|
|Storage:||-20±5℃||Applicable:||Micgene 244/244 IVD（BIO-KEY）|
High Risk HPV PCR Kit,
48 tests/kit HPV PCR Kit,
HPV PCR Kit Lyophilized
HPV Genotype 2+12 High risk(16+18)+Medium risk Real Time PCR Detection Kit (Lyophilized) --48 tests/kit
This kit, adopting Real-Time PCR-Fluorescence technology, is used for qualitative detection of 14 types of HPV(16, 18, 31, 33,
35, 39, 45, 51, 52, 56, 58, 59, 66, 68). It can be used for the laboratory diagnosis and monitoring of the virus infection. The test results are only for clinical reference, not for case confirmation or exclusion.
The specific primers and specific fluorescent probes are designed for the conserved region of HPV(16, 18, 31, 33, 35, 39,
45, 51, 52, 56, 58, 59, 66, 68). The PCR reaction solution, and real-time fluorescent quantitative PCR detection technology is applied on the fluorescence quantitative PCR instrument to realize the rapid detection of HPV through the change of fluorescence signal. The detection system contains a human internal control (IC) gene, which is used to monitor whether the specimens and the process of the nucleic acid extraction is qualified.
NG product lyophilization pacakage:
PCR Enzyme Mix (Lyophilized)
Note: Do not mix the components from different batches for detection. The Positive Control of HPV and internal control were constructed artificially, and they were not infectious.
1.Specimen type :
Cervical shedding cells, genitourinary tract secretions.
2.Specimen collection and transportation:
Clean the sampling site with sterile saline, and then use an epithelial brush (cervical brush, swab, etc.) to collect exfoliated cells from the cervix or genitourinary tract, epithelial cells at the lesion, secretions, etc. swabs, etc.) into a sterile storage tube and seal it for inspection.
The specimen can be stored for 2 weeks at 4-8℃, 3 months at -20±5℃ and for a long time below -70℃.
Negative Control: None of the FAM, VIC/HEX, ROX& Internal Control (CY5) channels have a Ct value or Ct ＞38;
Positive Control: FAM, VIC/HEX, ROX& Internal Control (CY5) channels Ct ≤ 30;
The above two items should be met simultaneously in one experiment. Otherwise, the experiment is invalid and should be repeated.
Data analysis and interpretation:
The following sample results are possible:
|Ct value||Result Analysis|
|3#||38~40||Re-test; if it is still 38~40, report as 2#|
Operation details please check from IFU!
Contact Person: Lisa