|Storage:||Store Away From Direct Light At -20±5℃||Applicable:||Micgene 242 / 244 、ABI QuantStudio5|
|Limit Of Detection:||3copies/μL||CV:||≤5%|
ISO13485 Vibrio Parahaemolyticus Rt Pcr Kit,
Vpa Pcr Amplification Kit,
Vpa Rt Pcr Kit Nucleic Acid
Vibrio parahaemolyticus kit Nucleic Acid PCR Detection Kit
(PCR-Fluorescent Probe Method)
This kit is suitable for the specific detection of pathogenic vibrio parahaemolyticus (Vpa) in bait, water samples, shrimp seedlings, shrimp tissues and processed products.
This kit uses a specific fragment of Vibrio parahaemolyticus (Vpa) to design a primer probe probe, which can specifically bind to a SECTION of DNA template in the middle of the primer amplification region. During the PCR extension process, the exonuclide activity of Taq enzyme cuts the 5 '-terminal fluorophore from the probe and makes it free in the reaction system. Thus, the 3 '-end quenched group was separated, and the fluorescence was emitted, which was detected by fluorescence PCR instrument to achieve the detection of Vibrio parahaemolyticus nucleic acid (Vpa) in the completely closed reaction system.
|PCR reaction liquid||24 tubes|
|Positive control||1 tube|
|Negative control||1 tube|
1. Sample type
Shrimp seedling, shrimp tissue, water body, bait, processed products, etc
2. Sample collection and transportation
Sample collection and transportation shall be carried out in accordance with NY/T 541. Samples shall be transported to the laboratory as soon as possible under refrigerated conditions to avoid repeated freeze-thaw.
3. Sample preservation
It can be stored for 24 hours at 2℃~8℃, and can be stored for a long time at -20℃.
Using Micgene 242/244/244IVD and ABI QuantStudio5:
procedures are set as below:
30sCollect the fluorescent
FAM was selected for detection channel.
The results of interpretation
|Negative||No Ct value or Ct value≥45|
Note: The result is judged to be grey area, which needs to be rechecked. If the recheck result is Ct value < 45 is considered as positive, and no Ct value or Ct value ≥45 is considered as negative.
Automatic analysis using instrumentation
Negative control: No obvious amplification curve of FAM detection channel;
Positive control: FAM channel showed obvious amplification curve, Ct value ≤30;
The above conditions are met simultaneously in the same experiment; otherwise, the experiment is invalid and re-testing is required.
Contact Person: Tracy